Featured product: NaviCross® Support Catheter
A 54-year-old diabetic patient presented with right foot ulcer. The CT angiography revealed long critical stenosis of the right EIA and a long SFA occlusion.
Arterial hypertension, IDDM, schizophrenia
Left retrograde CFA access with a 6 Fr crossover sheath was used to treat both lesions in one setting. The EIA was angioplastied with a 6x80mm PTA balloon catheter with good angiographic results. The sheath was advanced with the tip parked within the common femoral artery. A subintimal recanalization of the SFA was performed with a 0.035" Terumo Radifocus wire and was supported with a 5Fr straight catheter. The occlusion was crossed with a wire but the catheter would not track through the re-entry point due to lack of pushability. The catheter was exchanged for a 5x40mm PTA balloon catheter but attempts to track through the re-entry point were unsuccessful.
The PTA balloon catheter was exchanged for a NaviCross® which successfully tracked through the re-entry point into the lumen. The Terumo wire was exchanged for a 0.035" Bentson wire and the entire SFA was angioplastied to 6mm with significant recoil of both the entry and re-entry points. The sites which recoiled were stented with self-expanding stents with good results.
The NaviCross® support catheter has excellent trackability and pushability which are essential in difficult cases especially with contralateral up and over access.
This is HCP only content
website and cannot be refused. Other cookies are only placed if you choose to do so. For further information, please read our cookies policy.
Essential cookies to ensure the website functions well.
Cookies to help us measure the general use of our website.
Cookies that save personal information while browsing.